MiniMed 670G - To address and clear the alert or alarm
The audio/vibration pattern repeats every 5 minutes or every 15 minutes (depending on the alert) until the alert is cleared.
The audio/vibration pattern repeats every minute for 10 minutes if the alarm is not cleared. After 10 minutes, the alarm begins to siren
Read the text on the screen to understand the alert or alarm and the steps that should be taken
Press on the desired option.
Example of alarm:
It is important that you are able to address an Insulin flow blocked alarm.
This alarm means that insulin is not able to get through the tubing or cannula. If this alarm occurs, check your blood glucose and check to see if your infusion set has become dislodged or if your tubing is kinked.
If you don’t detect an issue and are unable to change your reservoir and infusion set right away, you might choose to select Resume Basal. If an Insulin flow blocked alarm occurs again, follow the steps on the screen. Select Rewind and change your reservoir and infusion set.
If you detect an issue or if your reservoir has run out of insulin, follow the steps on the screen. Select Rewind to change your reservoir and infusion set.
You can call the Product Helpline if you have questions about your pump, alerts or alarms.